current press releases
prior press releases
section 16 filings
Q. Is high resolution electrocardiography difficult or expensive to perform?
Q. What are late potentials?
Q. What are the principle clinical indications for high resolution electrocardiography?
Q. Can I get reimbursed for this procedure in the office?
Q. What is the clinical significance of late potentials?
Q. What should my clinical response be to the presence of late potentials in a patient?
Q. How are late potentials measured?
LP-Pac Q FAQs
Q1. After launching 'ACQUIRE' (from the LP-Pac Q™ acquisition menu), the dialog box opens with an error message that says "No QRS Detected".
Q2. After launching 'ACQUIRE' (from the LP-Pac Q™ acquisition menu) and entering LEAD CHECK, the graphics show one, two or three channels of flat-lines.
Q3. During lead check, it is observed that there is too much noise on one or more leads. What can I do?
1200 EPX FAQs
Q1. It seems to be taking a long time for patient file listings to appear. Is there anything I can do?
Q2. What should I do if I am getting a message that tells me to "Replace Internal Battery?"
Q3. I am having some trouble transferring PDATA files from the 1200EPX™ to my PC. Can you help?
Q4. Excessive noise is seen in the left panel of the 1200 EPX™ display, and it takes very long to perform an SAECG test. What should I do?
Micron Products, Inc.
The world's largest producer of high-performance one and two-piece Ag/AgCl disposable electrode sensors.
Copyright 2008 All Rights Reserved. Site designed by
Planteria Web Design